The best way to show this is in the form of an example. This also works in the same way for the DSP-100 and DSP-2000 CableAnalyzers.
To protect customers purchasing cabling systems, you cannot change the limit lines for NEXT, PSNEXT, ELFEXT, PSELFEXT, Attenuation and Return Loss. Any custom tests are clearly marked with an Asterisk, e.g. *Cat5*.
Пример: I want to test a link with only two pairs.
|Rotate the dial to SETUP.|
|Press Key 4 until you reach page 7|
|Decide on a name for your test; press ENTER to put the name into the analyzer.|
|Now we need to pick a standard to base the test on. In this example we will pick TIA Cat 3 Channel.|
|Now we need to change the cable name, in this example we shall enter SCTP.|
|Press Key 4 to advance to page 2. You can change the length an NVP here. Well keep it as it is in this example.|
|Press Key 4 will advance to page 3. Press ENTER to change.|
|Any configuration can be built.|
|Now press SAVE.|
|Now select the standard you just created. Go to page 1 of 7 and press ENTER.|
|Page down to find your custom limit and press ENTER. Once selected you can rotate the dial to AUTOTEST ready to go.|